We at Piercers.com wanted you to know that SciCan recommends a separate surge protector for your Statim y Bravo Autoclaves y Hydrim Washers.
SciCan Technical Service Bulletin
Bulletin number 18-TSB-387
Europa & International
Descripción: Canadá, Europa & International markets (not USA); Re: All HYDRSOY® Washer Disinfector units, C61WD and M2 G4—Surge Protector Recommendation.
This technical service bulletin is to recommend that a surge protector is used for the power connection when installing HYDRSOY C61WD y M2 G4 models. This is particularly important and highly recommended in areas where the voltage fluctuates in order to prevent damage to the unit’s internal components and maintain its functionality.
HYDRSOY models (rating listed below) are equipped with the power cord that is specific to the region it is sold into. Please refer to the appropriate unit’s operators manual for further details on the electrical requirements. You can use a secondary type surge protectors or power strips for this purpose. Please make sure not to connect any other equipment on the same outlet/power bar.
HYDRSOY M2 G4 – 230-240 VAC, 50Hz, 13A & 230-240 VAC, 60Hz, 13A
HYDRSOY C61 – 200-230 VAC, 50Hz, 12A & 208-240 VAC, 60Hz, 12A
If you have any additional questions please contact your local service department or SciCan representative.
SciCan Product Management